EPJ Web of Conferences
Volume 5, 2010API'09 - First NanoCharM Workshop on Advanced Polarimetric Instrumentation
|Number of page(s)||1|
|Published online||02 June 2010|
System and method for partial LCVR Stokes polarimeter thermal drift compensation
Laboratoire LAGIS, Université Lille 1,
In this paper we deal with the problem of thermal drift when using a LCVR (liquid crystal variable retarder) Stokes polarimeter. We address this problem in the restricted case of using a polarimeter to estimate partially linearly polarized light (that corresponds for example to the reflection of an unpolarized incident light on manufactured objects). Usually, the three parameters (S0, S1 and S2) are estimated by observing the reflected light wave, with a CCD sensor, through the LCVR polarimeter. The accuracy of the Stokes parameters estimate is then directly related to precise adjustment of the retardation. The problem is that the retardation introduced by a liquid crystal, variable retarder is strongly dependent on external factors (i.e. temperature). So, it is almost impossible to guarantee the reliability of the retardation in time and in practice it is either necessary to calibrate the LCVR just before carrying out measurement or use a thermostatic control. So, in this paper we propose a solution that does not care about the thermal drift of the LCVR to estimate accurately partially linearly polarized light without thermostatic control.
© Owned by the authors, published by EDP Sciences, 2010
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.