EPJ Web of Conferences
Volume 5, 2010API'09 - First NanoCharM Workshop on Advanced Polarimetric Instrumentation
|Number of page(s)||1|
|Section||New frontiers in polarimetry|
|Published online||02 June 2010|
Fast near infra-red ferroelectric liquid crystal based Mueller matrix system for imaging and spectroscopy
Department of Physics, Norwegian University of Science and
Technology, 7491, Trondheim, Norway
The science and optical engineering of imaging Mueller Matrix Ellipsometry (MME) and Spectroscopic MME is currently being revitalized based on an efficient optimal design method, and through the use of the so-called Eig envalue Calibration Technique (ECT). Through the ECT one may efficiently measure the details of the polarization state generator (PSG) matrix, and the polarization state analyzer (PSA) matrix, and hence avoid modeling of any unknown polarizing components in the system, and in particular the exact response of complex polarizing elements such as liquid crystal retarders. We here start up with presenting a detailed an alysis of the dynamic response of a near infrared Ferroelectric Liquid Crystal based Mueller matrix ellipsometer (NIR FLC- MME)  . A time dependent simulation model, using the measured time response of the individual FLCs, is used to describe the measured temporal response. Furthermore, the impulse response of the detector and the pre-amplifier is characterized and in cluded in the simulation model. The measured time dependent intensity response of the MME is well explained by simulations. A FLC based NIRMME system is here shown to be able to operate accurately at the maximum speed of approximately 16 ms per Mueller matrix measurement (steady state response). We demonstrate here time dependent Mueller Matrix measurements of a dynamically changing sample, with even down to 8 ms sampling time of each complete Mueller Matrix (with some loss of accuracy). We secondly briefly present the NIR-FLC- MME imaging system, and show applications to strain imaging of a crystal subjected to an external pressure. Furthermore, we present near-infra-red Mueller matrix images and corresponding polar decomposition images of thin slices of bio-tissue .
© Owned by the authors, published by EDP Sciences, 2010
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.