EPJ Web of Conferences
Volume 83, 2015QENS/WINS 2014 - 11th International Conference on Quasielastic Neutron Scattering and 6th International Workshop on Inelastic Neutron Spectrometers
|Number of page(s)||6|
|Published online||23 January 2015|
Recent progress on HYSPEC, and its polarization analysis capabilities
1 Oak Ridge National Laboratory, Quantum Condensed Matter Division, Oak Ridge, USA
2 Paul Scherrer Institut, Laboratory for Developments & Methods, Villigen PSI, Switzerland
3 Oak Ridge National Laboratory, Instrument and Source Division, Oak Ridge, USA
4 Oak Ridge National Laboratory, Neutron Data Analysis & Visualization, Oak Ridge, USA
5 European Spallation Source, Data Management and Software, Copenhagen, Denmark
6 Brookhaven National Laboratory, Condensed Matter Physics & Materials Science Department, Upton, USA
a Corresponding author: firstname.lastname@example.org
Published online: 23 January 2015
HYSPEC is a high-intensity, direct-geometry time-of-flight spectrometer at the Spallation Neutron Source, optimized for measurement of excitations in small single-crystal specimens with optional polarization analysis capabilities. The incident neutron beam is monochromated using a Fermi chopper with short, straight blades, and is then vertically focused by Bragg scattering onto the sample position by either a highly oriented pyrolitic graphite (unpolarized) or a Heusler (polarized) crystal array. Neutrons are detected by a bank of 3He tubes that can be positioned over a wide range of scattering angles about the sample axis. HYSPEC entered the user program in February 2013 for unpolarized experiments, and is already experiencing a vibrant research program. Polarization analysis will be accomplished by using the Heusler crystal array to polarize the incident beam, and either a 3He spin filter or a supermirror wide-angle polarization analyser to analyse the scattered beam. The 3He spin filter employs the spin-exchange optical pumping technique. A 60∘ wide angle 3He cell that matches the detector coverage will be used for polarization analysis. The polarized gas in the post-sample wide angle cell is designed to be periodically and automatically refreshed with an adjustable pressure of polarized gas, optically pumped in a separate cell and then transferred to the wide angle cell. The supermirror analyser has 960 supermirror polarizers distributed over 60∘, and has been characterized at the Swiss Spallation Neutron Source. The current status of the instrument and the development of its polarization analysis capabilities are presented.
© Owned by the authors, published by EDP Sciences, 2015
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.