EPJ Web Conf.
Volume 153, 2017ICRS-13 & RPSD-2016, 13th International Conference on Radiation Shielding & 19th Topical Meeting of the Radiation Protection and Shielding Division of the American Nuclear Society - 2016
|Number of page(s)||7|
|Section||7. Poster Presentations|
|Published online||25 September 2017|
Intrinsic noise of a superheated droplet detector for neutron background measurements in massively shielded facilities
1 Centro de Ciências e Tecnologias Nucleares, Instituto Superior Técnico, Universidade de Lisboa, 2695-066 Bobadela, Portugal
2 Laboratoire Souterrain à Bas Bruit, University of Nice, University of Avignon, Centre National de la Reserche Scientifique, Aix-Marseille University, Observatoire de la Côte d’Azur, 84400 Rustrel, France
3 Departamento de Física, Faculdade de Ciências, Universidade de Lisboa, 1749-016 Lisboa, Portugal
* Corresponding author: firstname.lastname@example.org
Published online: 25 September 2017
Superheated droplet detectors are a promising technique to the measurement of low-intensity neutron fields, as detectors can be rendered insensitive to minimum ionizing radiations. We report on the intrinsic neutron-induced signal of C2ClF5 devices fabricated by our group that originate from neutron- and alpha-emitting impurities in the detector constituents. The neutron background was calculated via Monte Carlo simulations using the MCNPX-PoliMi code in order to extract the recoil distributions following neutron interaction with the atoms of the superheated liquid. Various nuclear techniques were employed to characterise the detector materials with respect to source isotopes (238U, 232Th and 147Sm) for the normalisation of the simulations and also light elements (B, Li) having high (α, n) neutron production yields. We derived a background signal of ~10-3 cts/day in a 1 liter detector of 1-3 wt.% C2ClF5, corresponding to a detection limit in the order of 10-8 n cm-2s-1. Direct measurements in a massively shielded underground facility for dark matter search have confirmed this result. With the borosilicate detector containers found to be the dominant background source in current detectors, possibilities for further noise reduction by ~2 orders of magnitude based on selected container materials are discussed.
© The Authors, published by EDP Sciences, 2017
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.