EPJ Web Conf.
Volume 201, 2019The XXII International Scientific Conference of Young Scientists and Specialists (AYSS-2018)
|Number of page(s)||5|
|Published online||04 February 2019|
Calibration methods of new device for human blood pressure measurement
Wroclaw University of Science and Technology, Faculty of Mechanical and Power Engineering, Department of Criogenics, Aeronautics and Process Engineering, 50-370 Wroclaw, Wybrzeze Wyspianskiego 27, Poland
* Corresponding author: firstname.lastname@example.org
Published online: 4 February 2019
The paper presents two methods of determining calibration curve of a new device for blood pressure measurement. The device was developed at Wrocław University of Science and Technology. First method is based on parallel measurement of systolic and diastolic pressure measurement with use of reference device such as sphygmomanometer and researched new device with pneumatic sensor equipped with voltage type output. Obtained data (systolic ps and diastolic pd pressure, maximum us and minimum ud voltage) was then used to determine individual pressure-voltage characteristic of the device, which can be represented as a linear equation. Second method is based on substitution of experimentally proved coefficient b with its analytical equivalent extracted from mathematical model of described pneumatic sensor. Described methods were verified experimentally and compared. Metrological parameters of the device were designated.
© The Authors, published by EDP Sciences, 2018
This is an open access article distributed under the terms of the Creative Commons Attribution License 4.0 (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.