EPJ Web Conf.
Volume 225, 2020ANIMMA 2019 – Advancements in Nuclear Instrumentation Measurement Methods and their Applications
|Number of page(s)||6|
|Section||Nuclear Power Reactors Monitoring and Control|
|Published online||20 January 2020|
Test Results of an Innovative and Modular Sensor Design for Fission Chamber Studies
CEA, DEN, DER, Instrumentation, Sensors and Dosimetry Laboratory,
† CEA, DEN, DER, LP2E, Cadarache, F-13108 St-Paul-Lez-Durance, France
Published online: 20 January 2020
Since many years, the Instrumentation, Sensors and Dosimetry laboratory at CEA Cadarache has been working to propose innovative and robust design of fission chambers. In this framework, few simulation tools like Chester or PyFc were developed to make detector prototyping and optimisation possible. Up to now, the code experimental validations are scarce mainly because of the lack of flexibility of regular fission chambers. In order to set up an experimental database for code validation, an innovative modular twin fission chamber, the CFTM, was designed. It allows to set with precision the inter-electrode gap, the filling gas pressure as well as the fissile deposit used. This detector was irradiated in Minerve zero power reactor in order to gather results for code validation. Thanks to a calibration procedure, a bank of experimental results is available for simulation validation.
© The Authors, published by EDP Sciences, 2020
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.