EPJ Web Conf.
Volume 225, 2020ANIMMA 2019 – Advancements in Nuclear Instrumentation Measurement Methods and their Applications
|Number of page(s)||5|
|Published online||20 January 2020|
Usage of multiple fission cells for neutron flux measurements during rod-insertion method
Jožef Stefan Institute,
Published online: 20 January 2020
The measurements of physical parameters of the TRIGA reactor and Nuclear power plant Krško (NEK) reactor cores have been in the past performed on hand of the neutron flux signal obtained from uncompensated ionization cells and by employment of the a digital meter of reactivity (DMR). At the TRIGA reactor only one ionization cell is currently used for flux measurements. During the insertion of one control rod the neutron flux distribution is significantly altered affecting the flux measurements of inserting different control rods. The problem is presently solved by assigning a correction factor to each control rod what introduces an additional uncertainty.
In the present paper the implementation of four fission cells for reactivity measurements is presented. In this way determining the correct gamma background and its subtraction, performed by DMR algorithms, becomes less important as previously by using ionization chambers. The larger number of detectors also reduces the flux redistribution effects on the signal during individual control rod movements.
Key words: Fission cells / control rod worth measurements / research reactor / combining signals
© The Authors, published by EDP Sciences, 2020
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.