EPJ Web Conf.
Volume 225, 2020ANIMMA 2019 – Advancements in Nuclear Instrumentation Measurement Methods and their Applications
|Number of page(s)||10|
|Section||Environmental and Medical Sciences|
|Published online||31 January 2020|
Initial Performance Testing of Sri Gamma Spectroscopy Scintillators and Comparison to Other Improved-Resolution Detectors for Typical Health Physics Applications
Mirion Technologies − Canberra 800 Research Parkway
Published online: 31 January 2020
Europium-doped Strontium Iodide (SrI) scintillation gamma detectors have only recently become commercially available. SrI has very high light output ranging from 80,000-100,000 ph/MeV and energy resolution as good as 2.9% at 662 keV and 14.8% at 32 keV. The high density of 4.6 g/cc in combination with a high effective Atomic number and the absence of internal activity make SrI a very attractive sensor material for spectroscopic radiation monitoring devices. This new entrant into the field of commercially available gamma spectroscopy detectors offers improved energy resolution over NaI, without the elevated background of LaBr. While CZT and HPGe detectors have better resolution, the small available size of CZT and the cooling requirement of HPGe limit their usefulness in portable applications. Of these scintillators, SrI2 has the lowest detection limits when compared to other equal- size scintillation detectors, especially at low energies where U and Pu are of interest.
© The Authors, published by EDP Sciences, 2020
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.