EPJ Web Conf.
Volume 237, 2020The 29th International Laser Radar Conference (ILRC 29)
|Number of page(s)||4|
|Published online||07 July 2020|
ESA’s Space-Based Doppler Wind Lidar Mission Aeolus – First Wind and Aerosol Product Assessment Results
1 European Space Agency (ESA) ESTEC, Noordwijk, Netherlands
2 European Space Agency (ESA) ESRIN, Frascati, Italy
3 European Centre for Medium-Range Weather Forecasts (ECMWF), Reading, United Kingdom
4 Royal Dutch Meteorological Institute (KNMI), de Bilt, Netherlands
5 MétéoFrance, Toulouse, France
6 DoRIT, Oberpfaffenhofen, Germany
7 DLR, Institute of Atmospheric Physics, Oberpfaffenhofen, Germany
8 DLR, Remote Sensing Technology Institute, Oberpfaffenhofen, Germany
9 University of Applied Sciences, Munich, Germany
10 Ludwig-Maximillians-University, Meteorological Institute, Munich, Germany
11 LATMOS, Institut Pierre Simon Laplace (IPSL), Paris, France
* Email: Anne.Straume@esa.int
Published online: 7 July 2020
The European Space Agency (ESA) wind mission, Aeolus, hosts the first space-based Doppler Wind Lidar (DWL) world-wide. The primary mission objective is to demonstrate the DWL technique for measuring wind profiles from space, intended for assimilation in Numerical Weather Prediction (NWP) models. The wind observations will also be used to advance atmospheric dynamics research and for evaluation of climate models. Mission spin-off products are profiles of cloud and aerosol optical properties. Aeolus was launched on 22 August 2018, and the Atmospheric LAser Doppler INstrument (Aladin) instrument switch-on was completed with first high energy output in wind mode on 4 September 2018 , . The on-ground data processing facility worked excellent, allowing L2 product output in near-real-time from the start of the mission. First results from the wind profile product (L2B) assessment show that the winds are of very high quality, with random errors in the free Troposphere within (cloud/aerosol backscatter winds: 2.1 m/s) and larger (molecular backscatter winds: 4.3 m/s) than the requirements (2.5 m/s), but still allowing significant positive impact in first preliminary NWP impact experiments. The higher than expected random errors at the time of writing are amongst others due to a lower instrument out-and input photon budget than designed. The instrument calibration is working well, and some of the data processing steps are currently being refined to allow to fully correct instrument alignment related drifts and elevated detector dark currents causing biases in the first data product version. The optical properties spin-off product (L2A) is being compared e.g. to NWP model clouds, air quality model forecasts, and collocated ground-based observations. Features including optically thick and thin particle and hydrometeor layers are clearly identified and are being validated.
© The Authors, published by EDP Sciences, 2020
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.