EPJ Web of Conferences
Volume 119, 2016The 27th International Laser Radar Conference (ILRC 27)
|Number of page(s)||3|
|Section||Poster Session (Aerosol Observations and Retrievals II)|
|Published online||07 June 2016|
The System of the Calibration for Visibility Measurement Instrument Under the Atmospheric Aerosol Simulation Environment
1 Chinese Academy of Science, Key Laboratory of Geospace Environment, Hefei, Anhui, 230026, China
2 University of Science and Technology of China, School of Earth and Space Science, Hefei, Anhui, 230026, China
3 ShenZhen Darsunlaser Tech Co.,Ltd Shenzhen Guangdong, 518067, China
* Email: email@example.com
Published online: 7 June 2016
Visibility is one of the most important parameters for meteorological observation and numerical weather prediction (NWP).It is also an important factor in everyday life, mainly for surface and air traffic especially in the Aeronautical Meteorology. The visibility decides the taking off and landing of aircraft. If the airport visibility is lower than requirement for aircraft taking off stipulated by International Civil Aviation Administration, then the aircraft must be parked at the airport. So the accurate measurement of visibility is very important. Nowadays, many devices can be measured the visibility or meteorological optical range (MOR) such as Scatterometers, Transmissometers and visibility lidar. But there is not effective way to verify the accuracy of these devices expect the artificial visual method. We have developed a visibility testing system that can be calibration and verification these devices. The system consists of laser transmitter, optical chopper, phase-locking amplifier, the moving optic receiving system, signal detection and data acquisition system, atmospheric aerosol simulation chamber. All of them were placed in the atmosphere aerosol simulation chamber with uniform aerosol concentration. The Continuous wave laser, wavelength 550nm, has been transmitted into the collimation system then the laser beam expanded into 40mm diameter for compressing the laser divergence angle before modulated by optical chopper. The expanding beam transmitting in the atmosphere aerosol cabin received by the optic receiving system moving in the 50m length precision guide with 100mm optical aperture. The data of laser signal has been acquired by phase-locking amplifier every 5 meter range. So the 10 data points can be detected in the 50 meters guide once. The slope of the fitting curve can be obtained by linear fitting these data using the least square method. The laser extinction coefficient was calculated from the slope using the Koschmieder formula, then it been divided by 3 is MOR. The aerosol concentration in chamber can be changed by adjusting aerosol generator that producing variety of visibility atmospherical environment. The experiment has been carried out and the measurement accuracy of atmospheric transmittance is 0.3‰ Corresponding to the accuracy of MOR 4.9% at the 2km visibility environment. So this system can be calibrated and validated the other visibility measuring devices.
© Owned by the authors, published by EDP Sciences, 2016
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.