EPJ Web Conf.
Volume 146, 2017ND 2016: International Conference on Nuclear Data for Science and Technology
|Number of page(s)||4|
|Section||Experimental Facilities, Equipment, Techniques and Methods|
|Published online||13 September 2017|
A compact multi-plate fission chamber for the simultaneous measurement of 233U capture and fission cross-sections
1 CERN, Geneva, Switzerland
2 Irfu, CEA, Universiteí Paris-Saclay, Gif-sur-Yvette, France
3 CENBG, CNRS/IN2P3-Université de Bordeaux, Gradignan, France
4 CEA, DAM, DIF, 91297 Arpajon, France
5 EC-JRC Geel, Belgium
6 TUW Atominstitut, Vienna, Austria
a e-mail: Michael.Bacak@cern.ch
Published online: 13 September 2017
233U plays the essential role of fissile nucleus in the Th-U fuel cycle. A particularity of 233U is its small neutron capture cross-section which is about one order of magnitude lower than the fission cross-section on average. Therefore, the accuracy in the measurement of the 233U capture cross-section essentially relies on efficient capture-fission discrimination thus a combined setup of fission and γ-detectors is needed. At CERN n_TOF the Total Absorption Calorimeter (TAC) coupled with compact fission detectors is used. Previously used MicroMegas (MGAS) detectors showed significant γ-background issues above 100 eV coming from the copper mesh. A new measurement campaign of the 233U capture cross-section and alpha ratio is planned at the CERN n_TOF facility. For this measurement, a novel cylindrical multi ionization cell chamber was developed in order to provide a compact solution for 14 active targets read out by 8 anodes. Due to the high specific activity of 233U fast timing properties are required and achieved with the use of customized electronics and the very fast ionizing gas CF4 together with a high electric field strength. This paper describes the new fission chamber and the results of the first tests with neutrons at GELINA proving that it is suitable for the 233U measurement.
© The Authors, published by EDP Sciences, 2017
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.