EPJ Web Conf.
Volume 146, 2017ND 2016: International Conference on Nuclear Data for Science and Technology
|Number of page(s)||4|
|Section||Nuclear Masses, Structure and Decay Data Measurements|
|Published online||13 September 2017|
A low background ionisation chamber for alpha-spectroscopy
1 Institut für Kern- und Teilchenphysik, TU-Dresden, Dresden, Germany
2 Helmholtz Zentrum Dresden-Rossendorf (HZDR), Dresden, Germany
a e-mail: firstname.lastname@example.org
Published online: 13 September 2017
The goal of designing a low background ionisation chamber is to measure long lived α-decay half-lives which might interfere with rare event searches. Such decays play a part in many fields in nuclear physics and are difficult to measure. A lot of Geiger-Nutall studies also depend on them. Among others the research is specifically aimed at the precision measurements of α emitters mainly within the Lanthanide region. The excellent energy resolution would also allow to search for excited states in α-decays. To achieve this goal a gridded ionisation chamber was constructed using the Frisch-Grid design. A background rate of only 10.9(6) counts per day has been achieved in the energy region of 1 MeV to 9 MeV and improvements are possible. This low background rate and size of the chamber allows precision measurements of long living alpha decays with half-lives in the region of 1 × 1015 years.
© The Authors, published by EDP Sciences, 2017
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.