EPJ Web Conf.
Volume 147, 201719th Joint Workshop on Electron Cyclotron Emission and Electron Cyclotron Resonance Heating
|Number of page(s)||3|
|Published online||24 July 2017|
Testing of the Prototype Receiver for ITER ECE Diagnostic
1 ITER-India, Institute for Plasma Research, Bhat, Gandhinagar - 382428, India
2 University of Texas at Austin, Austin, TX, USA
* Corresponding author: firstname.lastname@example.org
Published online: 24 July 2017
The Electron Cyclotron Emission (ECE) diagnostic system in ITER provides essential information for plasma control and for evaluating the plasma performance. It measures the electron temperature profile (edge/core), electron temperature fluctuations and radiated power in the electron cyclotron frequency range from the plasma. Virginia Diodes Inc. (VDI) has designed and developed a state-of-the-art prototype receiver to detect emission in the 200-300 GHz frequency range. The prototype receiver developed by VDI has been tested at DIII-D tokamak to check its performance by measuring higher frequency ECE harmonics in high temperature plasmas. Bench testing verifies the receiver’s band coverage and noise temperatures. Good sensitivity has been obtained meeting the ITER requirement. Also, the receiver performance is assessed for ITER by comparing the data from the receiver and the existing Michelson interferometer for the same DIII-D plasma shot. The measurement results show that the receiver provides accurate temperature information for the plasma emission throughout the frequency band.
© The Authors, published by EDP Sciences, 2017
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.