EPJ Web Conf.
Volume 147, 201719th Joint Workshop on Electron Cyclotron Emission and Electron Cyclotron Resonance Heating
|Number of page(s)||6|
|Published online||24 July 2017|
Update on the status of the ITER ECE diagnostic design
1 Princeton Plasma Physics Laboratory, Princeton, NJ 08543, USA
2 Institute for Fusion Studies, University of Texas at Austin, TX 78712, USA
3 Center for Electromechanics, University of Texas at Austin, TX 78758, USA
4 ITER-India/Institute for Plasma Research, Bhat 382428, Gandhinagar, India
5 Plasma Science and Fusion Center, MIT, Cambridge, MA 02139, USA
6 ITER Organization, Route de Vinon sur Verdon, 13115, St Paul Lez Durance, France
a Corresponding author: email@example.com
Published online: 24 July 2017
Considerable progress has been made on the design of the ITER electron cyclotron emission (ECE) diagnostic over the past two years. Radial and oblique views are still included in the design in order to measure distortions in the electron momentum distribution, but the oblique view has been redirected to reduce stray millimeter radiation from the electron cyclotron heating system. A major challenge has been designing the 1000 K calibration sources and remotely activated mirrors located in the ECE diagnostic shield module (DSM) in the equatorial port plug #09. These critical systems are being modeled and prototypes are being developed. Providing adequate neutron shielding in the DSM while allowing sufficient space for optical components is also a significant challenge. Four 45-meter long low-loss transmission lines transport the 70–1000 GHz ECE from the DSM to the ECE instrumentation room. Prototype transmission lines are being tested, as are the polarization splitter modules that separate O-mode and X-mode polarized ECE. A highly integrated prototype 200–300 GHz radiometer is being tested on the DIII-D tokamak in the USA. Design activities also include integration of ECE signals into the ITER plasma control system and determining the hardware and software architecture needed to control and calibrate the ECE instruments.
© The Authors, published by EDP Sciences, 2017
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.