EPJ Web Conf.
Volume 170, 2018ANIMMA 2017 – Advancements in Nuclear Instrumentation Measurement Methods and their Applications
|Number of page(s)||8|
|Section||Safeguards, homeland security|
|Published online||10 January 2018|
Instrumentation effects on U and Pu CBNM standards spectra quality measured on a 500 mm3 CdZnTe and a 2×2 inch LaBr3 detectors
Nuclear Research Center SCK-CEN, B-2400 Mol, Belgium firstname.lastname@example.org
Nuclear Research Center SCK-CEN, B-2400 Mol, Belgium email@example.com; firstname.lastname@example.org; email@example.com
Université libre de Bruxelles, Servicede Métrologie Nucléaire (CP/165/84), firstname.lastname@example.org; email@example.com
European Commission Joint Research Center, Retieseweg 111, B-2440 Geel, Belgium Peter.SCHILLEBEECKX@ec.europa.eu
Published online: 10 January 2018
Nowadays, there is interest in developing gamma-ray measuring devices based on the room temperature operated medium resolution detectors such as semiconductor detectors of the CdZnTe type and scintillators of the LaBr3 type. This is true also for safeguards applications and the International Atomic Energy Agency (IAEA) has launched a project devoted to the assessment of medium resolution gamma-ray spectroscopy for the verification of the isotopic composition of U and Pu bearing samples. This project is carried out within the Non-Destructive Assay Working Group of the European Safeguards Research and Development Association (ESARDA). In this study we analyze medium resolution spectra of U and Pu standards with the aim to develop an isotopic composition determination algorithm, particularly suited for these types of detectors. We show how the peak shape of a CdZnTe detector is influenced by the instrumentation parameters. The experimental setup consisted of a 500 mm3 CdZnTe detector, a 2×2 inch LaBr3 detector, two types of measurement instrumentation – an analogue one and a digital one, and a set of certified samples – a 207Bi point source and U and Pu CBNM standards. The results of our measurements indicate that the lowest contribution to the peak asymmetry and thus the smallest impact on the resolution of the 500 mm3 CdZnTe detector was achieved with the digital MCA. Analysis of acquired spectra allowed to reject poor quality measurement runs and produce summed spectra files with the least impact of instrumentation instabilities. This work is preliminary to further studies concerning the development of an isotopic composition determination algorithm particularly suited for CZT and LaBr3 detectors for safeguards applications.
Key words: CdZnTe / LaBr3 / uranium enrichment / isotopic composition / medium resolution / safeguards
© The Authors, published by EDP Sciences, 2018
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited. (http://creativecommons.org/licenses/by/4.0/).
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.