EPJ Web Conf.
Volume 253, 2021ANIMMA 2021 – Advancements in Nuclear Instrumentation Measurement Methods and their Applications
|Number of page(s)||6|
|Section||Severe Accident Monitoring|
|Published online||19 November 2021|
Analysis of the Signal over Noise Ratio of the hodoscope determined by Monte Carlo calculation
IRSN/PSN-RES/SEREX/L2EP, Cadarache, Saint Paul Lez Durance 13115 FRANCE
Published online: 19 November 2021
The CABRI experimental pulse reactor, located at the Cadarache nuclear research center, southern France, is devoted to the study of Reactivity Initiated Accidents (RIA) for the purpose of the CABRI International Program (CIP), managed by IRSN in the framework of an OECD/NEA agreement. The hodoscope equipment installed in the CABRI reactor is an almost unique online fuel motion monitoring system, thanks to the measurement of the fast neutrons emitted during a power pulse by a tested rod positioned inside a dedicated test loop reproducing PWR conditions. This system is dedicated to the analysis of fuel displacement. Hence, one of the most important parameter measured by the hodoscope detectors is the Signal over Noise Ratio (SNR), characterizing the fraction of neutrons directly coming from the test rod (“signal”) over neutrons coming from the core (“noise”).
It is interesting to calculate the SNR in order to define some quantitative criterions to improve hodoscope measurements and to understand if any modification linked to the test loop may significantly change this essential parameter.
Another parameter of interest is the so-called “scattering coefficient”, which corresponds to the fraction of neutrons coming from the test rod and being scattered between their birth and their detection. This parameter is used to enhance the analysis of the fuel displacement which may happen during the power transient.
In this article, the method used to calculate the SNR using MCNP6.2 Monte Carlo code will be detailed. Because the hodoscope detectors are located far away from the test rod (up to 4 meters), a 2D model of CABRI core and instrumentation has been implemented. No variance reduction techniques have been used to solve this problem in order to record the place of birth of neutron which contributes to the different scores with the goal to perform a detailed analysis of the SNR. This strategy allows to access numerically to the “scattering coefficient”.
Finally, the comparison between calculated and measured SNR for a case study will be carried out. A quite good agreement between the 2D simulations and experiments recently performed in the CABRI reactor has been obtained.
Key words: CABRI / Hodoscope / MCNP / RIA / Neutron transport
© The Authors, published by EDP Sciences, 2021
This is an Open Access article distributed under the terms of the Creative Commons Attribution License 4.0, which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.