EPJ Web Conf.
Volume 203, 201920th Joint Workshop on Electron Cyclotron Emission and Electron Cyclotron Resonance Heating (EC20)
|Number of page(s)||4|
|Published online||25 March 2019|
Comparative studies of various types of transmission lines in the frequency range 70 GHz 1 THz for ITER ECE diagnostic
ITER-India, Institute for Plasma Research, Bhat, Gandhinagar - 382428, India
2 Homi Bhabha National Institute, Anushaktinagar, Mumbai - 400094, India
3 ITER Organization, Route de Vinon-sur-Verdon, CS 90 046, 13067 St. Paul Lez Durance Cedex, France
4 Princeton Plasma Physics Laboratory, Princeton, NJ 08543, USA
5 Institute for Fusion Studies, University of Texas at Austin, TX 78712, USA
* Corresponding author: email@example.com
Published online: 25 March 2019
In ITER, an Electron Cyclotron Emission (ECE) diagnostic is planned to measure the electron temperature by measuring the cyclotron radiation in the frequency range of 70-1000 GHz. The cyclotron radiation is usually of low power and needs to be transported with low attenuation over a long distance of ~ 43 m, through a suitable transmission system. Pertaining to long distance, the transmission system will consist of straight waveguide sections, miter bends and waveguide joints. Low power, low loss transmission in a broadband frequency range over long distance makes the design of the transmission system challenging. To arrive at a suitable transmission system, attenuation measurements of three types of transmission lines (TLs) have been performed i.e. circular smooth walled, corrugated and dielectric coated waveguide. A polarizing Michelson interferometer based on Martin-Puplett design has been used to measure the spectrum from waveguide set ups and liquid nitrogen has been used as the black body radiation source. The measured spectrum shows atmospheric water vapour absorption lines in all types of TLs. The preliminary measurement shows that the attenuation of smooth walled waveguide is found to be comparable to corrugated waveguide up to ~ 600GHz and better than corrugated waveguide above 600 GHz for the chosen set of experimental conditions. Further, to avoid water absorption lines, a smooth walled TL is evacuated up to rough vacuum (~10-2mbar) and it was observed that the attenuation is decreased and overall transmission is improved.
© The Authors, published by EDP Sciences, 2019
This is an open access article distributed under the terms of the Creative Commons Attribution License 4.0 (http://creativecommons.org/licenses/by/4.0), which permits unrestricted use, distribution, and reproduction in any medium, provided the original work is properly cited.
Current usage metrics show cumulative count of Article Views (full-text article views including HTML views, PDF and ePub downloads, according to the available data) and Abstracts Views on Vision4Press platform.
Data correspond to usage on the plateform after 2015. The current usage metrics is available 48-96 hours after online publication and is updated daily on week days.
Initial download of the metrics may take a while.